Payment & Shipping Terms:
|Cooling Method:||Peltier , Water-cooled||Cooling Capacity:||320W @ T-bath=25℃ T-facility=25℃|
|Power Supply:||94~265 VAC, Single Phase, 50/60HZ, 600W||Communications:||RC 3/8|
Semiconductor Peltier Bath Thermoelectric Tank Water Cooling
Principle of Peltier Device
A Peltier deivice (thermoelectric module, thermoelectric device) is a plate type element, inside which P-type semiconductors and N-Type semiconductors are located alter-nately. If direct current is supplied to the Peltier device , heat is transferred inside the device , and one face generates heat and increases temperature while the other face sucked heat and decreases temperature . Therefore, changing the direction of the current supplied to the Peltier device and achieve heating and cooling operation . This method has a fast response and can shift quickly between heating and cooling, so temperature can be controlled very precisely .
Prevention of electric shock and fire
Do not operate the switch with wet hands . Also , do not operate the Thermoelectric Bath with water or fluid left on it .
Action in the case of error
If any error such as abnormal sounds , smoke , or bad smell occurs , cut off the power at once , and stop supplying facility water . Please contact Adcol or a sales distributor to repaire the Themoelectric Bath .
The inspection must be done by an operator who has sufficient knowledge and experience.
a , Check of displayed contents
b , Check of temperature , vibration and abnormal sounds in the body of the Thermoelectric Bath .
c , Check of the voltage and current of the power supply system .
d , Check for leakage and contamination of the recirculating fluid and instrusion of foreign objects to it .
e , Check radiation air flow condition and temperature .
f , Check for leakage , quality change , flow rate and temperature of facility water .
|Control Method||Cooling/heating automatic PID control PID|
|Heat Radiation||Peltier element(Liquid-Cooled)|
|Operating Temp||0~60℃(for Water ≥4℃)|
|Cooling Capacity||320W @T-bath=25℃;T-facility=25℃|
|Heating Capacity||700W @T-bath=25℃;T-facility=25℃|
|Bath Liquid||Water : 4-60℃
Ethylene glycol-Water (≤50%) : 0-60℃
|Facility Liquid||Temp||10~35℃(No Dew Condensation)|
|Flow rate||2~7 L/min|
|Pressure||≤145 PSI (1 MPA)|
|port size||RC 3/8|
|Size||Bath||W216xD216xH250mm Usable Dept: 220mm||W300xD350xH400mm Usable Dept: 370mm|
|Drain Port size||CPC Coupling PLCD 16006|
|Power supply||94~265 VAC, Single phase, 50/60HZ, 600W|
|Over Current Protection||Rated current, 10A|
|Serial COMM||RS485( RS 232 Option)|
|Alarm Output||High/Low temp alarm, Output cutoff alarm|
|Temp. Sensor||PT100, 3 wires|
|Weight||Approx. 20KG||Approx. 30KG|